- Oral presentation
- Open Access
Lasers, a tool for soft cochleostomies
Head & Neck Oncology volume 2, Article number: O38 (2010)
Advancements in implantable auditory prostheses now demand preservation of residual auditory function following the surgery. Atraumatic cochleostomy formation is essential to this goal. Clinically reported hearing outcomes in human implantation are still quite variable in this regard. The objective of the study was to determine whether a CO2 laser operated with a handheld hollow waveguide can consistently produce cochleostomies without damaging the residual auditory function.
Human temporal bones were used to present a novel method creating a cochleostomy and adult guinea pigs were used to test whether cochlear function will be affected by the procedure. Baseline cochlear function was determined by recording compound action potential thresholds evoked by acoustic tone pips. Measurements were conducted at 6 steps per octave and 5 octaves starting at 50kHz. The sound level as attenuated from 0 to 80dB in steps of 5dB.
The human temporal bones were mounted and a standard approach through the facial recess was used to access the cochlea. The lateral bone was thinned at the basal turn with a motorized drill (Anspach Effort®, Palm Beach Gardens, FL, USA). Care was taken not to open the cochlea. The final opening of scala tympani was made with the handheld CO2 laser fiber (BeamPath® OTO-S, OmniGuide® Inc., Cambridge, MA, USA) and a Sharplan 20C CO2 laser (Lumenis®, Yokneam, Israel). In addition to describing the surgical approach, our experiments have demonstrated that for a careful selection of the laser's power, the safety range for the laser is superior to the safety range of drilling. Particularly important is the finding that multiple laser pulses through the same cochleostomy do not further increase the initial compound action potential (CAP) threshold elevation. Moreover, multiple laser pulses at different locations of the cochlea do not further increase the initial CAP threshold elevation observed after the first laser pulse. When opening the inner ear, safe laser settings would be up to 4W, at pulse durations of 100ms in single pulse mode.
About this article
Cite this article
Fishman, A., Moreno, L., Rivera, A. et al. Lasers, a tool for soft cochleostomies. Head Neck Oncol 2, O38 (2010). https://doi.org/10.1186/1758-3284-2-S1-O38
- Compound Action Potential
- Threshold Elevation
- Safety Range
- Hearing Outcome
- Cochlear Function